- Open Access
Development and implementation of explicit computerized protocols for mechanical ventilation in children
© Jouvet et al; licensee Springer. 2011
Received: 19 October 2011
Accepted: 21 December 2011
Published: 21 December 2011
Mechanical ventilation can be perceived as a treatment with a very narrow therapeutic window, i.e., highly efficient but with considerable side effects if not used properly and in a timely manner. Protocols and guidelines have been designed to make mechanical ventilation safer and protective for the lung. However, variable effects and low compliance with use of written protocols have been reported repeatedly. Use of explicit computerized protocols for mechanical ventilation might very soon become a "must." Several closed loop systems are already on the market, and preliminary studies are showing promising results in providing patients with good quality ventilation and eventually weaning them faster from the ventilator. The present paper defines explicit computerized protocols for mechanical ventilation, describes how these protocols are designed, and reports the ones that are available on the market for children.
Mechanical ventilation is a sophisticated technique that can keep alive the most severely ill patients; however, it can simultaneously damage the lungs and unfortunately generate unwanted complications . By analogy with pharmacology, mechanical ventilation can be viewed as a treatment with very narrow therapeutic windows, i.e., highly efficient but with considerable side effects if not used properly and in a timely manner. During the past two decades, considerable knowledge has been gained to find the optimal risk/benefit balance for mechanical ventilation. For instance, protective ventilation with low tidal volume (VT) and low airway pressure (Paw) has been shown to be safer than ventilation with high VT and Paw in adults with ARDS . However, several publications in adults and children have reported variable effects of written protocols in implementing protective ventilation with relatively low compliance with use of the protocols [3, 4] with a significant number of patients still being ventilated with high VT and high Paw [5, 6]. Human and organizational factors are at least partially responsible for such poor compliance  but so are the vast diversity of patient types, conditions, and changes over time, which makes one protocol unable to fit all situations.
In addition, expertise and human resources are not always available to make sure that patients receive the best ventilation everywhere and all the time. The ability to make timely adjustments of the ventilator according to the patient's condition, without much inter-caregiver variability, would certainly improve the safety and efficiency of mechanical ventilation especially when resources and expertise are not at the bedside 24 hours per day. In the present paper, we will define an explicit computerized protocol, describe how these protocols are designed and developed currently, report the ones that are available on the market for children, and propose some considerations for future developments in this field.
A protocol is a document that is designed to guide decisions regarding diagnosis, management, and treatment of specific medical situations. The protocol is based on the medical knowledge acquired from physiological studies, expertise, or evidence and can be generated by individuals or by consensus obtained from a group of physicians or experts. Protocols often are not precise enough to generate a decision at the bedside in a specific situation and thus significant inter-clinician variability in their application may exist.
An explicit protocol is designed to provide enough details to generate patient-specific therapy instructions that can be performed by different clinicians with no inter-clinician variability . Important individualization of patient therapy can be preserved by explicit protocols when they are driven by individual patient data. Considering the number of clinical situations and inputs from a given patient, explicit protocols rapidly become so complex that computers are required to integrate the large amount of information and provide specific answers to the user.
An explicit computerized protocol (ECP) is an explicit protocol supported by computer science to apply the instruction for a given patient at a given time. ECP might be in a laptop or integrated into the ventilator or monitoring device. The medical knowledge is usually implemented in the ECP through "if... then" rules. For example: if the SpO2 is < 88%, then increase FiO2 by 10%. The rules can be more complex and based on a validated physiological equation, such as the Otis equation  as used in IntelliVent™ (Hamilton, Bonaduz, Switzerland).
A rule can be in an open or closed loop. The rule is an open-loop rule if it results in a therapeutic or diagnostic recommendation displayed on the screen of a device for which caregivers can agree whether to accept the recommendation. According to the previous definitions, a clinical decision support system (CDSS) is defined as an ECP that uses two or more items of patient data to generate case-specific recommendations through rules that are only in open-loop .
Why do we need explicit computerized protocols for mechanical ventilation?
The human brain has a limited ability to incorporate data and information in decision making and human memory can simultaneously retain and optimally utilize only seven plus or minus two data constructs . The amount of data that is retained is even less when caregivers are working at night, with stress and/or time pressure. This limitation contrasts sharply with the clinical reality in which hundreds of variables are encountered by the caregivers in the ICU setting and decisions are made 24 hours per day. To prescribe mechanical ventilation, numerous parameters are considered, including all physiological data provided by the respirator, monitors, and clinical data from the charts on diagnosis and use of sedatives and hemodynamic treatments. The mismatch between human ability and the vast amount of data and information contributes to variation in clinical practice as decisions are made applying different data constructs and different knowledge/expertise. In such a complex environment, the help of an ECP is crucial to limit inter-caregiver variability. In a less complex environment, the aviation industry confronted human factors responsible for accidents  and decided decades ago to develop and implement closed-loop ECP in airplanes to improve safety resulting in a model of safety management today. That being said, in the medical field, the major limitation to developing ECP is to agree on which medical knowledge to implement.
Development of explicit computerized protocols
A multidisciplinary approach is needed to generate an ECP; the team should include clinical expert(s) in mechanical ventilation to generate the knowledge and validate ECP in a clinical environment, computer scientists to design the ECP platform, biomedical engineers to implement the ECP into medical devices (monitors and/or ventilators) and test ECP robustness and reliability, and industry to finalize a product that will receive a European Community marking (CE mark) and a U.S. marking (Food and Drug Administration (FDA)) approval.
Generation and validation of medical knowledge implemented in an ECP
The basic component of an ECP is a medical knowledge-based rule. In our clinical practice, we continuously apply rules. If we take the previous example of the SpO2/FiO2 rule, caregivers modify FiO2 according to SpO2 routinely. An ECP will recommend (open-loop) or do (closed-loop) in the same way, as soon as a valid SpO2 is available. The ECP also will define how often the FiO2 can be changed, the amplitude of change, and add additional rules: for example, define what will occur if FiO2 is 100% and SpO2 still below normal range.
The knowledge needed to develop an ECP able to manage the course of mechanical ventilation in any ICU patient is vast and we all know that "the devil is in the details." This knowledge is based on published work on respiratory physiology, clinical observational studies to describe current practice [6, 13], consensus conferences to define the specific clinical decision points (for example when do pediatric intensivists consider that we should switch from conventional mode to high-frequency ventilation mode in ARDS patients?) and clinical trials to validate ECPs [4, 14, 15]. A step-by-step approach, including more than one research center, is needed to develop valid, robust, and widely accepted ECPs. For example, the medical knowledge acquired in the past two decades on weaning in pressure support mode resulted in the development of SmartCare/PS™. SmartCare/PS™ was developed by one research team, and more than a decade elapsed from conception to commercialization [16, 17]. ECPs based on SpO2/FiO2 have already been developed for neonates and children and need further clinical validation [14, 15].
To shorten development and validation times, one option would be to follow the same development and validation process used in aviation by using a simulated flight environment (wind, temperature...). Medical ECP would need virtual patients with realistic physiological and pathological behaviors for developing and validating ECPs before clinical trials. Several teams are already working on such platforms, although none are currently commercialized for this purpose [18–20].
In the SpO2/FiO2 rule already described, the input data is patient SpO2, the control unit analyses SpO2 and selects the rule that corresponds to SpO2 value (e.g., increase FiO2 if SpO2 is low, decrease FiO2 if SpO2 is high (knowing that oxyhemoglobin dissociation curve is flat at SpO2 > 97%), or no change if SpO2 is in normal range), the output data is the FiO2 suggestion displayed on a screen (open-loop) or a setting modification on the respirator (closed-loop).
The input data, whether entered manually by the clinician or captured by a medical device, needs to be processed to discard artefacts and to be clinically relevant before being sent to the control unit . The key point is to input relevant, valid, robust, and stable data for the ECP. For example, the weaning ECP for children (SmartCare/PS™, Dräger, Germany) transforms real-time tidal volume, respiratory rate, and end tidal PCO2 (ETPCO2) into mean values during a 2-minute period. With IntelliVent™ (Hamilton Medical, Switzerland), the ETPCO2 used for the minute ventilation closed-loop is the second highest breath-by-breath ETPCO2 with enough quality index during the last 10 breaths . On the other hand, ETPCO2 is not always a good surrogate for PaCO2 especially during the acute phase of illness with high dead space. Such situations must be detected by the ECP to avoid any misinterpretation of the input data.
The control unit receives clinical information from the patient (input data) and "transforms" this information into orders (output data). The basic structure for processing information is rule-based (see above). All potential paths and situations should be addressed to lead to specific instructions. It usually requires several set of rules to manage two to three input data each. For example, a first set of rules could be for pressure support management according to tidal volume, respiratory rate, and ETPCO2 , a second set of rules could be for FiO2 and PEEP management according to SpO2 , and another set of rules could be for recommending extubation when pressure support, PEEP, and FiO2 reach threshold values for a certain length of time (mimicking an extubation readiness test) . Rules are usually "if... then ..." rules, but some ECPs have been developed using fuzzy logic , i.e., integrating patient's information in a fuzzy way to mimic the human brain [24, 25]. Despite the use of fuzzy logic in aircraft autopilot and in various other applications , to the best of our knowledge, fuzzy logic is not used in commercialized medical ECPs.
Output data can be recommendations to the caregivers suggesting new ventilator settings, specific orders such as "patient ready for separation from the ventilator," or ventilator settings being automatically adjusted. During the development phase of an ECP, output data are usually recommendations (open-loop). After extensive testing, some rules or sets of rules can be switched to close the loop.
A simple, attractively presented, and intuitive user interface is crucial to facilitate the understanding of the ECP decision process and for knowledge transfer at the bedside . Ideally, the user interface should include educational tools to train caregivers on mechanical ventilation management according to the ECP, as done with simulators for aircraft pilots.
As mentioned above, use of a virtual patient mimicking patient-ventilator cardiorespiratory interactions also is important for the development of ECPs. As for aircraft autopilot, a virtual patient may help "debug" the very first ECP versions but also aid understanding the complex interactions between rules and achieving preclinical validation. In addition such virtual patients, which should ideally be incorporated in the medical device (i.e., the ventilator), might be more efficient in training and teaching the eventual users (Figure 2). Currently, the virtual patients used for ECP development are computer simulations of the physiologic processes of respiration and circulation, using mathematical models.
Several barriers exist to the development of ECPs: 1) We do not generate enough medical knowledge in mechanical ventilation and most of the time ECPs are targeting a relatively small and regional scientific community. This can be improved by promoting multicenter international collaborations along the lines of the Pediatric Acute Lung Injury and Sepsis Investigators Network (PALISI), the European Society of Pediatric Intensive Care Medicine and Collaborative Critical Care Research Network (CCCRN) [5, 27]; 2) It is important to be able to capture any refusal of an ECP recommendation and to analyse if an adjustment of the ECP is mandatory. The commercialized ECP described below do not have such reporting systems. In the future, the ECPs should be equipped with a data report system and an expert team should analyze the data to refine the ECP. The versatility and the ease in upgrading and adjusting the ECP are probably key factors in making ECP widely accepted; 3) The manufacturers are unfortunately not ready to share their ECPs, processes and knowledge, for obvious marketing and business reasons. They also may need more resources and less time-to-market constraints to innovate further. Consortium(s) like those that exist in aeronautics could be of considerable value in driving forward innovation in this field.
Several barriers also exist to the acceptance and implementation of ECPs. These barriers include the lack of awareness, lack of familiarity with the protocol, lack of agreement, lack of demonstrated safety and efficacy, lack of known improved outcome, lack of ability to overcome the inertia of previous practice, protocol-related barriers (not easy to use, not convenient, cumbersome, confusing), environment-related barriers (new resources or facilities not accessible) . Among all these barriers, the safety issue is the first and most important one. The safety issue is addressed using the three following principles: 1) ECPs suggests a modification or modifies ventilation settings only within the alarms prescribed on the respirator; 2) there are stop rules implemented in ECPs that interrupt closed loop protocols in specific situations (if input data are not available for example); 3) Most of the ECPs in closed loop are first developed and tested for the management of the weaning phase. Despite these barriers many set of rules even in closed loop are widely accepted by caregivers. For example, the algorithm of the neurally adjusted ventilatory assist mode (NAVA) includes a rule that automatically adjusts positive inspiratory pressure to patient's electrical activity of the diaphragm change to deliver ventilation proportional to patient's needs.
Explicit computerized protocols for mechanical ventilation in children
Saxton and Myers reported the first ECP to adjust the end tidal PCO2 by regulating the negative pressure of an iron lung ventilator in poliomyelitis patients . Pediatricians then soon became concerned with adjusting the FiO2 tightly in order to avoid hypo/hyperoxemia and their related side effects [29, 30]. Dugdale and coworkers  reported in 1988 seven neonates with respiratory distress syndrome treated during 48 hours with a closed-loop FiO2 adjustment to keep the PaO2 obtained from an indwelling umbilical artery electrode at 10 kPa. The time spent by the neonates with PaO2 at ± 1 kPa of the target PaO2 was 75% with closed-loop control FiO2 compared with 45% with manual adjustment of FiO2. But the invasiveness of the PaO2 monitoring precluded the development of this ECP. Claure et al. conducted a research program designed to develop closed-loop FiO2 adjustment using SpO2 [32, 33]. These authors recently conducted a multicenter, randomized, clinical trial that included mechanically ventilated preterm infants who were ventilated during two consecutive 24-hour periods: one with FiO2 adjusted by caregivers and the other by an automated system, in random sequence. Automated FiO2 adjustment improved maintenance of the intended SpO2 range, and led, significantly, to reduced time with high SpO2 and more frequent episodes with SpO2 between 80% and 86% . This ECP is now commercialized as CLiO2™ (CareFusion, Yorba Linda, US) and implemented in a respirator.
Characteristics of the SmartCare/PS™ Draeger Medical and IntelliVent™ Hamilton Medical explicit computerized protocols
Type of breath
PC and PS
Body weight range for use
Primary goal of the ECP
wean while maintaining
ETPCO2, RR, Vt
within predefined range
ETPCO2, RR, Vt, SpO2
within predefined ranges
IBW, humidification system, medical history
IBW, medical history
Clinical decision support Option (open-loop)
2 minETPCO2, 2 minRR, 2 minVt, PEEP, PS level,
ETPCO2, RR, Vt, SpO2 PEEP, PIP, breath by breath
MV, PEEP, FiO2
Recommendation for separation from the ventilator
SBT duration displayed
IntelliVent™ (Hamilton Medical, Bonaduz, Switzerland) is an ECP for the automated control of minute volume, PEEP, and FiO2 in adaptive support ventilation (ASV). IntelliVent™ manages, with (open-loop) or without the need of caregivers intervention (closed-loop), the four following weaning steps: 1) switch from control ventilation to spontaneous breathing (specific to ASV mode); 2) automatic adaptation of the pressure to maintain the patient in a range of respiratory rate, end tidal PCO2 and SpO2; 3) an automated SBT when children reach minimum ventilation support; and 4) a timer that shows SBT duration (Table 1). IntelliVent™ is available on the G5 and S1 generation ventilators from Hamilton Medical that continuously monitor usual mechanical ventilation parameters plus ETPCO2 and SpO2 . IntelliVent™ has been assessed in one clinical trial on feasibility and safety in children during the weaning phase. Fifteen children were included and IntelliVent™ was safe and kept patients with body weight ≥7 kg in the "zone of respiratory comfort" comparably to PSV or ASV . The major strength of IntelliVent™ is the combination of an ECP and user-friendly interface that allows a certain individual customization of the protocols, including automatic recruitment maneuvers. There are several improvements to consider for IntelliVent™ in children: 1) at present, there is not much clinical experience with ASV and this ECP in children; 2) children with ideal body weight < 7 kg are excluded, therefore, another ECP is needed in the same PICU for younger children; 3) automatic PEEP adjustment needs additional validation; 4) IntelliVent limits adjustments of PEEP in the case of hemodynamic instability based on the presence of pulsus paradoxus on pulse oximetry waveform (heart lung index) [37, 38]. The accuracy of the waveform assessment and the upper limits of PEEP adjustment need to be validated in children.
New ECP developments
Considering the current context in the PICU (manpower shortages, increased complexity of care, lack of specific knowledge, needs for quality, safety, and reproducibility), ECP for mechanical ventilation will very soon be a "must." Open-loop and decision support systems allow too much room for inter-physicians variability and may not achieve the goal of providing the best possible care. As in many other fields, safety and quality will be achieved by closing the loop but allowing physicians to retain the role of supervisor, most directly involved with the most severe or atypical patients. Several closed-loop systems are already on the market, and preliminary studies have shown promising results in providing patients with good quality ventilation and eventually weaning them faster from the ventilator.
The authors thank Catherine Farrell, MD, for help with the manuscript preparation.
- Tobin MJ: Advances in mechanical ventilation. NEJM 2001, 344: 1986–1996. 10.1056/NEJM200106283442606PubMedView ArticleGoogle Scholar
- Putensen C, Theuerkauf N, Zinserling J, Wrigge H, Pelosi P: Meta-analysis: ventilation strategies and outcomes of the acute respiratory distress syndrome and acute lung injury. Ann Intern Med 2009, 151: 566–576.PubMedView ArticleGoogle Scholar
- Randolph AG, Wypij D, Venkataraman ST, Hanson JH, Gedeit RG, Meert KL, et al.: Effect of mechanical ventilator weaning protocols on respiratory outcomes in infants and children: a randomized controlled trial. JAMA 2002, 288: 2561–2568. 10.1001/jama.288.20.2561PubMedView ArticleGoogle Scholar
- Jouvet P, Farges C, Hatzakis G, Monir A, Lesage F, Dupic L, et al.: Weaning children from mechanical ventilation with a computer-driven system (closed-loop protocol): a pilot study. Pediatr Crit Care Med 2007, 8: 425–432. 10.1097/01.PCC.0000282157.77811.F9PubMedView ArticleGoogle Scholar
- Santschi M, Jouvet P, Leclerc F, Gauvin F, Newth CJ, Carroll CL, et al.: Acute lung injury in children: therapeutic practice and feasibility of international clinical trials. Pediatr Crit Care Med 2010, 11: 681–689. 10.1097/PCC.0b013e3181d904c0PubMedView ArticleGoogle Scholar
- Khemani RG, Sward K, Morris A, Dean JM, Newth CJ: Variability in usual care mechanical ventilation for pediatric acute lung injury: the potential benefit of a lung protective computer protocol. Intensive Care Med 2011, in press.Google Scholar
- Rubenfeld GD: Implementing effective ventilator practice at the bedside. Curr Opin Crit Care 2004, 10: 33–39. 10.1097/00075198-200402000-00006PubMedView ArticleGoogle Scholar
- Morris AH: Developing and implementing computerized protocols for standardization of clinical decisions. Ann Intern Med 2000,132(5):373–383.PubMedView ArticleGoogle Scholar
- Otis A, Fenn W, Rahn H: Mechanics of breathing in man. J Appl Physiol 1950, 2: 592–607.PubMedGoogle Scholar
- Wyatt J: Computer-based knowledge systems. Lancet 1991, 338: 1431–1436. 10.1016/0140-6736(91)92731-GPubMedView ArticleGoogle Scholar
- Miller G: The magical number seven plus or minus two: some limits on our capacity for processing information. Psychol Rev 1956, 63: 81–97.PubMedView ArticleGoogle Scholar
- Higton P: Safety lessons from aviation. Perfusion 2005, 20: 191–193. 10.1191/0267659105pf805oaPubMedView ArticleGoogle Scholar
- Eddington A, Jouvet P, Bordessoule A, Emeriaud G, Bruschschwein F, Wysocki M: Invasive versus non-invasive criteria ventilator setting modification in critically ill children. Pediatr Crit Care Med 2011,12(Suppl 3):A78.Google Scholar
- Jouvet P, Eddington A, Bordessoule A, Emeriaud G, Lopez Gasco R, Wysocki M: Weaning children from mechanical ventilation with an explicit computerized protocol: Intellivent ® . Pediatr Crit Care Med 2011,12(Suppl 3):A38.Google Scholar
- Claure N, Bancalari E, D'Ugard C, Nelin L, Stein M, Ramanathan R, et al.: Multicenter crossover study of automated control of inspired oxygen in ventilated preterm infants. Pediatrics 2011, 127: e76-e83. 10.1542/peds.2010-0939PubMedView ArticleGoogle Scholar
- Lellouche F, Mancebo J, Jolliet P, Roeseler J, Schortgen F, Dojat M, et al.: A multicenter randomized trial of computer-driven protocolized weaning from mechanical ventilation. Am J Respir Crit Care Med 2006, 174: 894–900. 10.1164/rccm.200511-1780OCPubMedView ArticleGoogle Scholar
- Dojat M, Brochard L, Lemaire F, Harf A: A knowledge-based system for assisted ventilation of patients in intensive care units. Int J Clin Monit Comput 1992, 9: 239–250. 10.1007/BF01133619PubMedView ArticleGoogle Scholar
- Sailors RM, East TD: A model-based simulator for testing rule-based decision support systems for mechanical ventilation of ARDS patients. Proc Annu Symp Comput Appl Med Care 1994, 1007.Google Scholar
- Winkler T, Krause A, Kaiser S: Simulation of mechanical respiration using a multicompartment model for ventilation mechanics and gas exchange. Int J Clin Monit Comput 1995, 12: 231–239. 10.1007/BF01207204PubMedView ArticleGoogle Scholar
- Jouvet P: Un simulateur reproduit les réactions d'un patient pour l'assistance respiratoire. Vecteur 2011, 10: 1.Google Scholar
- Tehrani FT, Roum JH: Intelligent decision support systems for mechanical ventilation. Artif Intell Med 2008, 44: 171–182. 10.1016/j.artmed.2008.07.006PubMedView ArticleGoogle Scholar
- Lellouche F, Hernert P, Jouvet P, Bouchard P, Brochard L: Clinical evaluation of a computer-driven algorithm to automatically set PEEP and FiO2 during mechanical ventilation. Am J Respir Crit Care Med 2009, 179: A3652.Google Scholar
- Olliver S, Davis GM, Hatzakis GE: Weaning infants with respiratory syncytial virus from mechanical ventilation through a fuzzy-logic controller. AMIA Annu Symp Proc 2003, 499–503.Google Scholar
- Bates JH, Young MP: Applying fuzzy logic to medical decision making in the intensive care unit. Am J Respir Crit Care Med 2003, 167: 948–952. 10.1164/rccm.200207-777CPPubMedView ArticleGoogle Scholar
- Jamshidi M: Tools for intelligent control: fuzzy controllers, neural networks and genetic algorithms. Philos Transact A Math Phys Eng Sci 2003, 361: 1781–1808. 10.1098/rsta.2003.1225View ArticleGoogle Scholar
- Wysocki M, Brunner JX: Closed-loop ventilation: an emerging standard of care? Crit Care Clin 2007, 23: 223–240. ix 10.1016/j.ccc.2006.12.011PubMedView ArticleGoogle Scholar
- Khemani RG, Newth CJ: The design of future pediatric mechanical ventilation trials for acute lung injury. Am J Respir Crit Care Med 2010,182(12):1465–1474. 10.1164/rccm.201004-0606CIPubMed CentralPubMedView ArticleGoogle Scholar
- Saxton GA, Myers G: A servomechanism for automatic regulation of pulmonary ventilation. J Appl Physiol 1957, 11: 326–328.PubMedGoogle Scholar
- Beran AV, Taylor WF, Ackerman BD, Sperling DR, Strauss J: An automatic oxygen control system for infants. Pediatrics 1971, 48: 315–318.PubMedGoogle Scholar
- Beddis IR, Collins P, Levy NM, Godfrey S, Silverman M: New technique for servo-control of arterial oxygen tension in preterm infants. Arch Dis Child 1979, 54: 278–280. 10.1136/adc.54.4.278PubMed CentralPubMedView ArticleGoogle Scholar
- Dugdale RE, Cameron RG, Lealman GT: Closed-loop control of the partial pressure of arterial oxygen in neonates. Clin Phys Physiol Meas 1988, 9: 291–305. 10.1088/0143-0815/9/4/001PubMedView ArticleGoogle Scholar
- Claure N, Gerhardt T, Everett R, Musante G, Herrera C, Bancalari E: Closed-loop controlled inspired oxygen concentration for mechanically ventilated very low birth weight infants with frequent episodes of hypoxemia. Pediatrics 2001, 107: 1120–1124. 10.1542/peds.107.5.1120PubMedView ArticleGoogle Scholar
- Claure N, D'Ugard C, Bancalari E: Automated adjustment of inspired oxygen in preterm infants with frequent fluctuations in oxygenation: a pilot clinical trial. J Pediatr 2009, 155: 640–645. e1–2 10.1016/j.jpeds.2009.04.057PubMedView ArticleGoogle Scholar
- Santschi M, Gauvin F, Hatzakis G, Lacroix J, Jouvet P: Acceptable respiratory physiologic limits for children during weaning from mechanical ventilation. Intensive Care Med 2007, 33: 319–325. 10.1007/s00134-006-0414-0PubMedView ArticleGoogle Scholar
- Jouvet P, Payen V, Gauvin F, Santschi M, Eddington A, Lacroix J: Weaning children from mechanical ventilation with a computer-driven explicit protocol: a randomized clinical trial. Am J Respir Crit Care Med 2010, 181: A3899.Google Scholar
- Rose L, Presneill JJ, Johnston L, Cade JF: A randomised, controlled trial of conventional versus automated weaning from mechanical ventilation using SmartCare/PS. Intensive Care Med 2008, 34: 1788–1795. 10.1007/s00134-008-1179-4PubMedView ArticleGoogle Scholar
- Michard F, Chemla D, Richard C, Wysocki M, Pinsky MR, Lecarpentier Y, et al.: Clinical use of respiratory changes in arterial pulse pressure to monitor the hemodynamic effects of PEEP. Am J Respir Crit Care Med 1999, 159: 935–939.PubMedView ArticleGoogle Scholar
- Feissel M, Teboul JL, Merlani P, Badie J, Faller JP, Bendjelid K: Plethysmographic dynamic indices predict fluid responsiveness in septic ventilated patients. Intens Care Med 2007, 33: 993–999. 10.1007/s00134-007-0602-6View ArticleGoogle Scholar
- Jouvet P, Mersmann S, Wysocki M: Automation of weaning in children. In Textbook on Pediatric and Neonatal Mechanical Ventilation. Edited by: Rimensberger P. New York: Springer Verlag; 2012:in press.Google Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.