- Open Access
Effect of high-frequency oscillatory ventilation on esophageal and transpulmonary pressures in moderate-to-severe acute respiratory distress syndrome
Annals of Intensive Care volume 6, Article number: 84 (2016)
High-frequency oscillatory ventilation (HFOV) has not been shown to be beneficial in the management of moderate-to-severe acute respiratory distress syndrome (ARDS). There is uncertainty about the actual pressure applied into the lung during HFOV. We therefore performed a study to compare the transpulmonary pressure (P L) during conventional mechanical ventilation (CMV) and different levels of mean airway pressure (mPaw) during HFOV.
This is a prospective randomized crossover study in a university teaching hospital. An esophageal balloon catheter was used to measure esophageal pressures (Pes) at end inspiration and end expiration and to calculate P L. Measurements were taken during ventilation with CMV (CMVpre) after which patients were switched to HFOV with three 1-h different levels of mPaw set at +5, +10 and +15 cm H2O above the mean airway pressure measured during CMV. Patients were thereafter switched back to CMV (CMVpost).
Ten patients with moderate-to-severe ARDS were included. We demonstrated a linear increase in Pes and P L with the increase in mPaw during HFOV. Contrary to CMV, P L was always positive during HFOV whatever the level of mPaw applied but not associated with improvement in oxygenation. We found significant correlations between mPaw and Pes.
HFOV with high level of mPaw increases transpulmonary pressures without improvement in oxygenation.
Moderate or severe acute respiratory distress syndrome (ARDS)  is associated with substantial mortality. Use of a lung-protective strategy with low tidal volume (V t) of 6 ml/kg of predicted body weight has been associated with improved outcomes . High-frequency oscillatory ventilation (HFOV) is a non-conventional mode which has been proposed to achieve the targets of protective ventilation with very low V t  and a greater alveolar stability due to relatively constant mean airway pressure (mPaw) . However, two large recently published randomized clinical trials, OSCAR  and OSCILLATE , failed to prove any clinical benefit when HFOV was applied in adults with moderate-to-severe ARDS as compared with a strategy with low tidal volume, high positive expiratory pressure (PEEP) and limited plateau pressure (Pplat). In the latter study, side effects of HFOV were observed with more requirements for vasopressors, likely due to right ventricular failure secondary to high mPaw used [7, 8].
Another possible explanation of the lack of clinical benefit with HFOV in adults with ARDS may be due to the occurrence of pulmonary overdistension in non-dependant areas of the lung . Because mPaw during HFOV does not reflect of the real pressure applied to the alveoli , with non-predictable attenuation all along the trachea–bronchial tree, it is not possible to know the true pulmonary distending pressure. Esophageal pressure (Pes) is an approximation of the pleural pressure, and its use has shown a possible clinical benefit when PEEP was set according to the value of Pes in moderate-to-severe ARDS . Esophageal pressure measurement allows the calculation of the maximal and minimal transpulmonary pressures (P L) applied during mechanical ventilation. Data reporting P L during HFOV are scarce  and only describe the feasibility of the technique but not the comparison of range of P L occurring during the switch from CMV to an HFOV trial. Therefore, we performed a prospective study of P L determination in moderate-to-severe ARDS during and after an HFOV trial.
This is an ancillary study of a previously published study .
The study was approved by the ethics committee of the Marseille University Hospital (Comité de Protection des Personnes Sud Méditerranée, ID RCB:2008-A00077-48). Written informed consent was obtained from each patient’s next of kin. Patients admitted in the intensive care unit of a university teaching hospital during a 10-month period were screened if they met inclusion criteria: moderate-to-severe ARDS with a PaO2/FiO2 ratio ≤150 mmHg at a PEEP ≥8 cm H2O. Exclusion criteria were age <18 years, moribund status, risks associated with HFOV (head injury, pneumothorax or a chest tube in place with persistent air leak) and contraindications to the placement of a nasogastric probe. All patients were sedated and continuously paralyzed . The severity of illness was determined according to the Simplified Acute Physiologic II Score, the Sepsis-related Organ Failure Assessment Score and the Lung Injury Score [14, 15].
Tested ventilatory strategies
Patients were submitted to a 6-h period of CMV (CMVpre) in volume-controlled, constant square flow, mode using the AVEA ventilator (VIASYS Healthcare, Palm Springs, CA, USA) with a tidal volume of 6 mL/kg of predicted body weight adjusted to obtain a plateau pressure <30 cm H2O. PEEP and FiO2 were adjusted according to the ARMA protocol . Patients were then switched to HFOV using a 3100B ventilator (SensorMedics, Yorba Linda, CA, USA). After a recruitment maneuver was performed with a mPaw of 40 cm H2O during 40 s with a pressure amplitude of oscillation of 0 cm H2O , HFOV was set as follows: FiO2 as during the CMVpre period; frequency of 5 Hz; inspiratory time of 33 %; and bias flow of 40 L/min. The pressure amplitude of oscillation and frequency were then adjusted to achieve a PaCO2 close to the PaCO2 measured during the CMVpre period. If pressure amplitude of oscillation of 110 cm H2O was insufficient to achieve a pH ≥ 7.25, frequency was decreased at 4 Hz. The protocol consisted of three 1-h periods of HFOV (HFO + 5, HFO + 10, HFO + 15) in a randomized order, with a mPaw level calculated by adding 5, 10 or 15 cm H2O to the mPaw measured during the CMVpre period (Fig. 1). A recruitment maneuver was performed at the beginning of each HFOV period and before switch back to CMV. Respiratory frequency and pressure amplitude of oscillation were adjusted to maintain PaCO2 constant during the protocol. Measurements were taken at the end of each period of the protocol and 1 h after switch back to CMV. During the protocol, norepinephrine infusion was adjusted to maintain a mean arterial pressure above 65 mmHg.
Esophageal and transpulmonary pressure measurements
A specific nasogastric feeding probe (SmartCath®, VIASYS Healthcare, Palm Springs, CA, USA) equipped with an esophageal balloon was inserted after in vitro automatized test for leak search and compliance measurement, and then the balloon was filled with a volume of air between 0.5 and 2 mL as recommended by the manufacturer. Every 30 min, the ventilator evacuates and refills the balloon to maintain measurement accuracy. The correct positioning in the lower third of the esophagus was confirmed by the presence of cardiac artifacts, the changes in transpulmonary pressure during tidal ventilation and the parallelism of airway and esophageal curves after the interruption of a brief chest compression maneuver . Finally, a chest X-ray excluded the misplacement of the probe into the airway. Esophageal pressures were recorded and monitored by the integrated system, CP-100 pulmonary monitor (Bicore Monitoring System Inc®, Irvine, CA, USA) present in the AVEA ventilator. An end-inspiratory occlusion of 2 s allowed the measurement of, respectively, Pplat and inspiratory Pes (Pesinsp), whereas an end-expiratory occlusion of 5 s allowed the measurement of, respectively, total PEEP (PEEPtot) and expiratory Pes (Pesexp) during CMV. During HFOV periods, because interruption of ventilation is not possible, screen of the AVEA ventilator was frozen for measuring the peak and trough amplitude of oscillations for measurements of, respectively, the maximum and minimum Pes. The following formulas were computed as follows:
An example of tracings in the two ventilatory modes with the airway, esophageal and transpulmonary pressures determinations and calculations is provided in Fig. 2.
Data are presented as mean ± SD or median (interquartile range) as required. Normality of variables was tested according the Kolmogorov–Smirnov test. Repeated-measures analysis of variance or Friedman’s test was used to evaluate the effect of time and mPaw level. The Tukey test or the Wilcoxon test was used for intergroup comparisons. Bivariate correlations with Spearman’s test for each period of ventilation were performed. All statistics and figures were performed with the SPSS 20.0 package (SPSS, Chicago, IL, USA).
Among the 16 patients included in the princeps study , ten patients were monitored by the esophageal catheter and were used in this study.
Table 1 reports patient’s characteristics at inclusion. Initial computed tomography scan or thoracic radiograph showed five lobar and five diffuse presentations. Causes of ARDS were bacterial pneumonia (n = 4), influenza A (H1N1), (n = 2), aspiration (n = 2), post-cardiopulmonary resuscitation (CPR) (n = 1) and acute pancreatitis (n = 1). Fluid loading was performed in three patients during the CMVpre period. At baseline, all except one received norepinephrine infusion to maintain mean arterial pressure (MAP) above 65 mmHg.
During HFOV, mPaw was progressively increased from 18 ± 4 cm H2O in CMVpre period to 33 ± 4 cm H2O at HFO + 15 (Table 2). PaO2/FiO2 ratio did not significantly change under HFOV when compared with the CMVpre period. However, it increased by more than 20 % in three patients at HFO + 5, in four patients at HFO + 10 and in two patients at HFO + 15. FiO2 was slightly lower at the end of the study. Worsening of oxygenation occurred in two patients at HFO + 5, in three patients at HFO + 10 and in four patients at HFO + 15. As required by the protocol, PaCO2 and pH were kept constant throughout the study. Concerning respiratory mechanics, Pplat, driving pressure, respiratory system elastance, chest wall elastance and pulmonary elastance were similar during the CMVpre and CMVpost periods. These last parameters could not be calculated during the HFOV periods because of the lack of tidal volume monitoring.
At similar level of mPaw, during the volumetric periods of ventilation (CMVpre and CMV post), we did not find differences concerning esophageal and transpulmonary pressures. During HFOV periods, we observed a linear increase in esophageal pressure from 12.4 [10.6; 16.7] to 19.1 [16.7; 23.3] cm of H2O (p = 0.001) as mPaw increases from 23 ± 4 to 33 ± 4 cm of H2O. As a consequence, mean P L increased during HFOV periods from 10.5 [7.3; 13.8] to 14 [11.5; 16.3] cm of H2O (Table 2; Fig. 3). Interestingly, there was no negative transpulmonary pressure whatever the period of HFOV ventilation. There were, however, seven (out of ten) patients with negative minimal P L during the CMVpre period and only three (out of ten) patients during the CMVpost period (p = 0.07, χ 2 test).
During HFO, mPaw was correlated with Pesmean at HFO + 5 and HFO +15 periods (respectively, ρ = 0.71, p = 0.02 and ρ = 0.84, p = 0.02) but at no time with P Lmean.
The present study assessing the use of esophageal pressure measurements in patient with moderate-to-severe ARDS on whom a trial of HFOV is performed demonstrates (1) a linear increase in transpulmonary pressures with the increase in mPaw during HFOV, (2) a minimal transpulmonary pressure which was always >0 during HFOV and (3) a correlation between mean esophageal pressure and mPaw.
For decades, HFOV has been used for respiratory failure in both adults and children who were inadequately responsive to conventional mechanical ventilation. However, recently the results of the OSCAR  and OSCILLATE  studies performed on adults have not shown benefit to HFOV over conventional ventilation. A recent study in the pediatric population has also shown equivocal results with HFOV . Indeed, positive studies on HFOV are limited [16, 19] and predate the era of low tidal volume conventional mechanical ventilation. During HFOV, there is uncertainty about the real pressure that is applied to the alveoli and therefore the distending pressure applied into the lung. Henderson et al.  have previously described the use of esophageal manometry to measure P L during HFOV. With a mean airway pressure of 27 ± 5 cm H2O during HFOV, they measured a mean esophageal pressure of 14 ± 4 cm H2O and computed a mean P L of 18 ± 4 cm H2O. These data are consistent with the present results, namely a mPaw of 28 ± 4 cm H2O, results in a median of 16.7 IQR [12.5; 18.7] cm H2O range of Pes and a median of 13.1 IQR [9.2; 14.7] cm H2O range of P L. The safe range of P L during HFO is not known. However, during conventional mechanical ventilation for ARDS, a P L > 27 cm H2O is associated with an unacceptably high level of strain . The P L value recorded during HFO remains below this threshold whatever the level of mPaw.
One interesting result is the correlation between mPaw and esophageal pressure that we obtained; the more mPaw is set, the more Pes is measured. In clinical practice, levels of mPaw in the OSCAR and OSCILLATE trials [5, 6] were not exactly the same. During the first 2 days of the studies, mPaw was set at 5 cm H2O higher in the Canadian trial than in the UK trial. These differences could have led to more pulmonary overdistension and side effects that could explain the deleterious outcomes observed with HFOV in the OSCILLATE trial.
An ongoing study, the EPOCH study , which aim is to compare a strategy of preventing atelectrauma with a P L of 0 cm H2O at end expiration to a strategy of lung recruitment to target P L of 15 cm H2O at end-inspiratory volume in a crossover design either with CMV and either with HFOV will clarify the protective or deleterious roles of HFOV as compared to CMV.
First, as measurements of esophageal pressure could not be taken in static conditions during HFO periods, we cannot rule out a possible bias of measurements due to cardiac artifacts. However, the use of mean esophageal pressure reduces this bias. Second, during HFOV, due to the lack of V t monitoring, we use the calculation of P L derived from Pes measurements  and not the elastance-derived measurements of P L  which could lead to different results . Indeed, experimental data have shown that although recorded value of Pes is a quite accurate approximation of measured pleural pressure in the middle part of the lungs, Pes can overestimate or underestimate the value of pleural pressure whether in the non-dependant part and whether in the dependant part of the lungs . The more convincing results are that the variations of Peso reflect those in pleural pressure whatever the parts of the lung . There is still a matter of controversy on the use of the former or the latter method. A prospective ongoing study could bring a response to the clinical utility of the method used . Third, because we have not performed the registration of airway pressure during HFO, we cannot exclude negative P L during the active expiratory phase, and further studies are needed to conclude. And fourth, from a technical point of view, we also cannot exclude that larger inflation volume as demonstrated by Mojoli et al.  could have led to different results. However, our study precedes the one from Mojoli, and we used the volume and the proceeding recommended by the manufacturer.
We cannot speculate whether lower mPaw during HFOV, the same range as recorded in CMV, could lead to lower esophageal and transpulmonary pressures recorded. A level of <+10 cm H2O of mPaw during HFO does not increase significantly Peso and P L. Only a level of ≥+15 cm H2O of mPaw increases significantly both Peso and P L.
The use of high mean airway pressures during HFOV leads to increase in transpulmonary pressures. Contrary to CMV, during HFOV, transpulmonary pressure remains always positive.
ARDS Definition Task Force, Ranieri VM, Rubenfeld GD, Thompson BT, Ferguson ND, Caldwell E, et al. Acute respiratory distress syndrome: the Berlin Definition. JAMA. 2012;307:2526–33.
Ventilation with lower tidal volumes as compared with traditional tidal volumes for acute lung injury and the acute respiratory distress syndrome. The Acute Respiratory Distress Syndrome Network. N Engl J Med. 2000;342:1301–08.
Hager DN, Fuld M, Kaczka DW, Fessler HE, Brower RG, Simon BA. Four methods of measuring tidal volume during high-frequency oscillatory ventilation. Crit Care Med. 2006;34:751–7.
McCulloch PR, Forkert PG, Froese AB. Lung volume maintenance prevents lung injury during high frequency oscillatory ventilation in surfactant-deficient rabbits. Am Rev Respir Dis. 1988;137:1185–92.
Young D, Lamb SE, Shah S, MacKenzie I, Tunnicliffe W, Lall R, et al. High-frequency oscillation for acute respiratory distress syndrome. N Engl J Med. 2013;368:806–13.
Ferguson ND, Cook DJ, Guyatt GH, Mehta S, Hand L, Austin P, et al. High-frequency oscillation in early acute respiratory distress syndrome. N Engl J Med. 2013;368:795–805.
Guervilly C, Forel JM, Hraiech S, Demory D, Allardet-Servent J, Adda M, et al. Right ventricular function during high-frequency oscillatory ventilation in adults with acute respiratory distress syndrome. Crit Care Med. 2012;40:1539–45.
Ursulet L, Roussiaux A, Belcour D, Ferdynus C, Gauzere BA, Vandroux D, et al. Right over left ventricular end-diastolic area relevance to predict hemodynamic intolerance of high-frequency oscillatory ventilation in patients with severe ARDS. Ann Intensive Care. 2015;5:25. doi:10.1186/s13613-015-0068-6.
Dreyfuss D, Ricard JD, Gaudry S. Did studies on HFOV fail to improve ARDS survival because they did not decrease VILI? On the potential validity of a physiological concept enounced several decades ago. Intensive Care Med. 2015;41:2076–786.
Hirayama T, Nagano O, Shiba N, Yumoto T, Sato K, Terado M, et al. Mean lung pressure during adult high-frequency oscillatory ventilation: an experimental study using a lung model. Acta Med Okayama. 2014;68:323–9.
Talmor D, Sarge T, Malhotra A, O’Donnell CR, Ritz R, Lisbon A, et al. Mechanical ventilation guided by esophageal pressure in acute lung injury. N Engl J Med. 2008;359:2095–104.
Henderson WR, Dominelli PB, Griesdale DE, Talmor D, Sheel AW. Airway pressure and transpulmonary pressure during high-frequency oscillation for acute respiratory distress syndrome. Can Respir J. 2014;21:107–11.
Papazian L, Forel JM, Gacouin A, Penot-Ragon C, Perrin G, Loundou A, et al. Neuromuscular blockers in early acute respiratory distress syndrome. N Engl J Med. 2010;363:1107–3116.
Le Gall JR, Lemeshow S, Saulnier F. A new Simplified Acute Physiology Score (SAPS II) based on a European/North American multicenter study. JAMA. 1993;270:2957–63.
Vincent JL, Moreno R, Takala J, Willatts S, De Mendonça A, Bruining H, et al. The SOFA (Sepsis-related Organ Failure Assessment) score to describe organ dysfunction/failure. On behalf of the Working Group on Sepsis-Related Problems of the European Society of Intensive Care Medicine. Intensive Care Med. 1996;22:707–10.
Ferguson ND, Chiche JD, Kacmarek RM, Hallett DC, Mehta S, Findlay GP, et al. Combining high-frequency oscillatory ventilation and recruitment maneuvers in adults with early acute respiratory distress syndrome: the Treatment with Oscillation and an Open Lung Strategy (TOOLS) Trial pilot study. Crit Care Med. 2005;33:479–86.
Higgs BD, Behrakis PK, Bevan DR, Milic-Emili J. Measurement of pleural pressure with esophageal balloon in anesthetized humans. Anesthesiology. 1983;59:340–3.
Bateman ST, Borasino S, Asaro LA, Cheifetz IM, Diane S, Wypij D, et al. Early high-frequency oscillatory ventilation in pediatric acute respiratory failure: a propensity score analysis. Am J Respir Crit Care Med. 2016;193:495–503. doi:10.1164/rccm.201507-1381OC.
Derdak S, Mehta S, Stewart TE, Smith T, Rogers M, Buchman TG, et al. High-frequency oscillatory ventilation for acute respiratory distress syndrome in adults: a randomized, controlled trial. Am J Respir Crit Care Med. 2002;166:801–8.
Chiumello D, Carlesso E, Cadringher P, Caironi P, Valenza F, Polli F, et al. Lung stress and strain during mechanical ventilation for acute respiratory distress syndrome. Am J Respir Crit Care Med. 2008;178:346–55. doi:10.1164/rccm.200710-1589OC.
Knowles JH, Hong SK, Rahn H. Possible errors using esophageal balloon in determination of pressure-volume characteristics of the lung and thoracic cage. J Appl Physiol. 1959;14:525–30.
Gattinoni L, Chiumello D, Carlesso E, Valenza F. Bench-to-bedside review: chest wall elastance in acute lung injury/acute respiratory distress syndrome patients. Crit Care. 2004;8:350–855.
Gulati G, Novero A, Loring SH, Talmor D. Pleural pressure and optimal positive end-expiratory pressure based on esophageal pressure versus chest wall elastance: incompatible results. Crit Care Med. 2013;41:1951–7.
Pelosi P, Goldner M, McKibben A, Adams A, Eccher G, Caironi P, et al. Recruitment and derecruitment during acute respiratory failure: an experimental study. Am J Respir Crit Care Med. 2001;164:122–30.
Chiumello D, Cressoni M, Colombo A, Babini G, Brioni M, Crimella F, et al. The assessment of transpulmonary pressure in mechanically ventilated ARDS patients. Intensive Care Med. 2014;40:1670–8.
Mojoli F, Chiumello D, Pozzi M, Algieri I, Bianzina S, Luoni S, et al. Esophageal pressure measurements under different conditions of intrathoracic pressure. An in vitro study of second generation balloon catheters. Minerva Anestesiol. 2015;81:855–64.
CG, JMF and LP designed the study. CG and JMF coordinated the study. CG, JMF, SH, AR and LP were responsible for patient enrolment and measurements of esophageal pressures. CG and JMF performed statistical analysis. CG, JMF, DT and LP analysed the data and wrote the manuscript. All authors read and approved the final manuscript.
The authors declare that they have no competing interests.
About this article
Cite this article
Guervilly, C., Forel, JM., Hraiech, S. et al. Effect of high-frequency oscillatory ventilation on esophageal and transpulmonary pressures in moderate-to-severe acute respiratory distress syndrome. Ann. Intensive Care 6, 84 (2016). https://doi.org/10.1186/s13613-016-0181-1
- Acute Respiratory Distress Syndrome
- Conventional Mechanical Ventilation
- Transpulmonary Pressure
- Esophageal Pressure
- Pleural Pressure